Wholesale Price China Digital Sensor - JIRS-BA-S800 Blue and Green Algae Sensor – JIRS Detail:
|Size||Diameter 37mm* Length 220mm|
|Main Materials||Main Body: SUS316L+ PVCO type Ring: FluororubberCable: PVC|
|Measurement Range||100-300,000 cells/mL|
|Measuring accuracy||±5% of the corresponding value of the 1 ppb rhodamine WT dyesignal level|
|Calibration||Deviation Calibration, Slope Calibration|
|Cable Length||Standard 10-Meter Cable, Max Length: 100 Meters|
|Warranty Period||1 Year|
|Working conditions||The distribution of blue-green algae in water is very uneven. It is recommended to monitor more than one point; the water turbidity is lower than 50NTU.|
2.1 Product Information
The blue-green algae sensor utilizes the characteristic that the cyanobacteria has an absorption peak and an emission peak in the spectrum. The spectral absorption peak of the cyanobacteria emits monochromatic light to the water, and the cyanobacteria in the water absorbs the energy of the monochromatic light, releasing another wavelength. With monochromatic light emitting peaks, the intensity of light emitted by cyanobacteria is proportional to the amount of cyanobacteria in water. The sensor is easy to install and use. Widely used in monitoring of blue-green algae in water stations, surface water, etc. The sensor is shown in Figure 1.
Figure 1 Blue-green Algae Sensor Appearance
3.1 Installation of Sensors
The specific installation steps are as follows:
a. Install the 8 (mounting plate) on the railing by the pool with 1 (M8 U-shape clamp) at the sensor mounting position;
b. Connect 9 (adapter) to 2 (DN32) PVC pipe by glue, pass the sensor cable through PVC pipe until the sensor screws into 9 (adapter), and do waterproof treatment;
c. Fix 2 (DN32 tube) onto 8 (mounting plate) by 4 (DN42U-shape clamp).
Figure 2 Schematic Diagram on the Installation of Sensor
|1-M8U-shape Clamp（DN60）||2- DN32 Pipe (outside diameter 40mm)|
|3- Hexagon Socket Screw M6*120||4-DN42U-shape Pipe Clip|
|5- M8 Gasket（8*16*1）||6- M8 Gasket（8*24*2）|
|7- M8 Spring Shim||8- Mounting Plate|
|9-Adaptor(Thread to Straight-through)|
3.2 Connection of Sensor
The sensor should be correctly connected by the following definition of wire core:
|Signal||+12VDC||AGND||RS485 A||RS485 B|
Product detail pictures:
Related Product Guide:
Bear "Customer initially, High-quality first" in mind, we work closely with our prospects and supply them with efficient and specialist companies for Wholesale Price China Digital Sensor - JIRS-BA-S800 Blue and Green Algae Sensor – JIRS , The product will supply to all over the world, such as: Portugal, Cape Town, Austria, Our expert engineering team will generally be prepared to serve you for consultation and feedback. We are able to also offer you with free of charge samples to meet your requirements. Best efforts will likely be produced to provide you the best service and merchandise. When you are keen on our business and products, please speak to us by sending us emails or call us quickly. In an effort to know our products and company extra, you may come to our factory to view it. We will generally welcome guests from all over the world to our business to create business relations with us. Please feel cost-free to speak to us for small business and we believe we'll share the best trading experience with all our merchants.
By Frank from Slovenia - 2018.02.12 14:52
The goods are very perfect and the company sales manager is warmful, we will come to this company to purchase next time.
By Sandy from Mumbai - 2018.12.28 15:18